See a log of previous and current versions describing the changes and improvements of each version.
ABOUT FIRMWARE VERSION 3.31 [Manuals version 1.5]
TAGARNO firmware version is available starting November 30, 2018 and contains the following corrected inexpedient behaviors discovered in release 3.30.
- By mistake the PNG image format was also being compressed, leading to a capture time of up to 15 seconds per image. This additional time does not make up for the benefits of a smaller PNG image and uncompressed PNG format is now selected as default.
- A slip in version 3.30 caused the adjustment of the Optimal working distance to take unacceptably long time. This has now been corrected.
- After updating the microscope to 3.30, the colors observed by the camera was over saturated and incorrect until the microscope was power cycled. This firmware update initializes all camera settings at startup and does not require the user to perform a power cycle after upgrading.
ABOUT FIRMWARE VERSION 3.30 [Manuals version 1.5]
TAGARNO firmware version is available starting November 1, 2018 and contains the following additions and corrections:
1. New options in the Settings Menu
- System languages
- 中文 (Chinese)
- 한국의 (Korean)
2. Lock presets
When this feature is enabled, users are prevented from accidentally overwriting presets in the Preset manager.
- Changes to the Preset manager
- Each preset can be assigned a user defined name by pressing F2 on an attached keyboard.
- The number of presets has been upgraded from four to six presets.
- Prevent overwriting a prefix configuration by enabling Lock presets in the Settings Menu.
- The tiff Image format has been changed from uncompressed to LZW compressed. The LZW compression reduces the size of the captured image dynamically, where an uncompressed tiff image always would take up 8.101 KB of storage space. LZW compression is lossless and does not degrade the quality of the captured image.
- When performing a calibration in TMA Auto exposure mode was always activated when the user selected New calibration. It has proven not to be appropriate in all situations and now Auto exposure mode has been added as a user selectable button on the calibration dialog.
- In TMA the LED light can now be switched on/off by long pressing the light key on the control box.
- The default Monitor width has been changed from 478mm to 531mm to match the size of a standard 24” monitor
- Manually changing the Monitor width while Monitor auto detect is enabled will now disable the monitor auto detect function.
- Some new monitors and televisions would not display an image when connected to the microscope. This issue has been solved by automatically changing the video format from DVI to HDMI depending on the capabilities of the attached monitor.
- In the Image comparison application the reference image was not always shown when selecting a template. In side-by-side view the sample image was not always shown. These intermittent issues have now been corrected.
- When Wide dynamic range was selected and the user entered Manual exposure mode the settings for Iris, Gain and Exposure time could be incorrect and result in a dark image. This issue has been fixed by allowing the camera to settle for a few seconds before it enters Manual exposure mode.
- Restore factory default settings did not delete the Image comparison templates. This has now been corrected.
ABOUT FIRMWARE VERSION 3.20 [Manuals version 1.4]
TAGARNO firmware version is available starting August 1, 2018 and contains the following additions and corrections:
1. File share mode
- Updated to support an SMB 2.0 compatible protocol.
This firmware update contains a security update with regards to using File share mode. The SMB 1.0 protocol was previously supported but it has been terminated because it posed a security risk. The security risk does not apply to the TAGARNO FHD device, but to any Windows based operating system having SMB 1.0 file sharing support enabled. It is highly recommended to disable SMB 1.0 support on any Windows based operating system on your corporate network.
2. An incompatible USB storage can now be formatted
- This TAGARNO FHD device is designed to support USB storages with a FAT32 formatted file system. Other file systems (like APFS, NTFS and exFAT) are not supported.
- If an incompatible USB storage is detected selecting Menu/Information/Storage information will display this message The USB storage must be formatted before it can be used.
- A USB storage can be formatted by selecting Menu//Format USB storage.
- If an incompatible USB storage is detected when capturing an image the user will be prompted to allow the TAGARNO FHD device to format the attached USB storage.
- When performing an angle measurement in TMA, Reflex Angle notation was used for angles greater than 90 degrees. This has now been corrected to only use Reflex Angle notation for angles greater than 180 degrees.
- Fixed an issue where a double cursor initially was shown in TMA before the mouse was moved.
ABOUT FIRMWARE VERSION 3.10 [Manuals version 1.3]
TAGARNO firmware version is available starting May 1, 2018 and contains the following additions and corrections:
1. New option in the Camera menu
- High contrast. Default off. Depending of the nature of the visual inspection type, High contrast can be activated to visually enhance the edges of microscopic specimens.
2. New option in the Settings Menu
- Image preview. When an image is captured the default behavior is to preview the image at the center of the display for 5 seconds. Image preview can now be disabled, which will display a more discrete text box for 3 seconds. This allows users to quickly inspect and capture images of new samples.
- Auto exposure mode has been changed to prioritize the Exposure time which is optimal for the selected video format. In good lightening conditions the microscope will reduce the Iris instead of the Exposure time to improve the depth of view.
- Manual focus mode has been changed to improve speed and precision. In Manual focus mode the initial focus speed is lowered but it will accelerate when - or + is depressed.
3. New features in TMA
- Short pressing the Snapshot key on the control box now saves the current image including any measurements and annotations.
- A warning dialog now prevents users from deleting a calibration unintentionally.
- Warning dialogs now supports input from an attached keyboard (ESC, Enter and left/right arrow)
4. New Image comparison application
ABOUT FIRMWARE VERSION 3.00 [Manuals version 1.2]
TAGARNO firmware version is available starting February 1, 2018 and contains the following additions and corrections:
1. New features in Focus stacking
Normal. Default fast mode with a typical processing speed of 8 seconds for 15 images
High. Reduce visual artifacts caused by sub optimal lighting and high objects. In this mode the processing speed is reduced and 15 images can take up to 13 seconds
- All settings used during Focus stacking, except zoom level, are persisted during power down
- The actual magnification level is displayed inside the Zoom key
- The control box zoom +/- keys have been activated as an alternative to using the onscreen Zoom key
- The LED light can now be switched on/off by long pressing the light key on the control box
- Long pressing the center key on the control box will now close the Focus stacking application or cancel an ongoing focus stacking
- The actively selected function is now more clearly marked with a highlighted border
- Fixed an issue where the application occasionally could close just after activation
2. New features in TMA
- Added Diameter measurement tool
- Auto completion has been added to the text annotation field allowing users to easily select previous entered annotations
- Function key shortcuts (F1 – F10) have been assigned to the drawing and annotation functions. The tooltip shows the assigned function key
- Color settings, line width and text size are now persisted during power down
- When Clear All is selected a warning dialog now enables users to cancel the operation if it was selected by accident
- In Polygon measurement the presentation of the result has been changed from Area = xx.x2 to A = xx.x2
- Clicking the magnification field now auto selects the current zoom level allowing users to easily enter a new zoom level using the keyboard
- During distance measuring, hold down the Shift key to measure only in vertical or horizontal direction
3. New options in the Settings Menu
- System languages: Polish and Japanese
- Keyboard layout: PT (Portuguese)
4. New option in the Camera menu
- Wide dynamic range. When selected the camera will use individual exposure settings for dark and light areas. This option can be used to reduce glare when inspecting blank objects. Please note that Wide dynamic range is not supported in Manual exposure mode.
ABOUT FIRMWARE VERSION 2.80 [Manuals version 1.1.6]
TAGARNO firmware version 2.80 contains the following additions and corrections:
1. New Watermark application
- Imprint captured images with Image prefix, time stamp, hotizontal field of view, microscope model and serial number
- Import and add a custom defined watermark on captured image
2. New Verification lines application
- Insert lines vertically and/or horizontally with a user defined number of lines either baseline or continued. State the distance between the lines and mark the tolerance
- Insert a grid with a user defined number of lines and state the wanted distance between the lines
3. New feature in TMA
- Polyline measurement. Measure the distance between two or more points
4. New options in the setup menu:
- Image prefix: Replace the default image prefix (img_) preceding the date and time stamp in filenames of captured images
5. Added system languages
- Italiano (Italian)
- Maymar (Hungarian)
6. Added keyboard layouts
- HU (Hungarian)
- RO (Romania)
- An issue in the embedded USB 3.0 live streaming device could result in a frozen image on the attached PC/Laptop. This firmware release addresses this issue and can maintain a live image on a PC/Laptop having a configuration below the otherwise recommended specifications.
- Fixed a minor issue where capturing an image with graphics would shortly show an empty filename in the window displaying the captured image.
ABOUT FIRMWARE VERSION 2.70 [Manuals version 1.1.5]
TAGARNO firmware version 2.70 contains the following additions and corrections:
1. New Application shortcut is added, bringing you directly to your fovaorite application via the control box
2. Spanish is added as an optional system language
3. Correction of a minor issue, so that the displayed magnification level is now updated correctly after warm restart
ABOUT FIRMWARE VERSION 2.51 [Manuals version 1.1.4]
TAGARNO firmware version 2.51 contains:
1. Optimization of the internal filesystem access
- Microscope power-up time has been reduced to less than 30 seconds
- Read/Write speed in file share mode has been increased by a factor 10
2. Added support in the Settings Menu:
- System Language: Pусский (Russian)
- Keyboard layout: RU (Cyrillic)
3. Changed feature in the Info Menu
- Restore factory default settings will now perform a quick format of the Internal storage.
All images captured by the user will be deleted. Calibrations performed in the Measurement application will also deleted. The user is prompted before Restore factory default settings is invoked.
ABOUT FIRMWARE VERSION 2.40 [Manuals version 1.1.2]
TAGARNO Firmware version 2.40 contains the following additions and corrections:
1. File share mode: Store images directly to your microscope and share the content of the microscopes internal storage directly from a computer via the activated LAN connection
2. New system languages: French, German and Danish. This choice affects both menus and applications
3. Numeric formats. Select between the numeric formats 1,234.56, 1.234,56 or 1 234,56
4. Changes in the graphical user interface:
- To increase readability and the amount of visible text, the use of block capitals in the Graphical user interface has been terminated
- Graphical user interface elements have gone through a brush up resulting in smoother edges and transparency adjustments
- Wherever possible, language-agnostic icons have replaced text in the Graphical user interface
5. General improvements:
- When an image is captured, the preview, filename and destination is displayed for a few seconds
- Comprehensive tooltips have been added to TMA. They appear at mouseover over each functionality
- The video format can now be changed without power cycling the microscope
- When performing an optimum working distance adjustment, the magnification settings prior to the adjustment will now be restored.
- The timestamp used in auto generated file names of captured and saved images included milliseconds. Milliseconds have now been omitted
- Storage information has been added in the Information menu, showing internal and USB device capacity
- TMA has been added mouse sensitivity in 5 steps
- Performing an optimal working distance adjustment while lock magnification was active would not work as expected
- At video format frame rates less than 1080p60 the focus stacking application could generate an image being partly out of focus
- At large magnifications, the scale on the vertical ruler could be missing the least significant digits
- Plugging in a USB storage device could cause an attached USB mouse/keyboard to stop working
- In auto capture mode an issue could result in one or more images being skipped
- A calibration in the measurement application using sheet I could fail. The calibration routine has been made less sensitive to ink residue and imperfect dots. The calibration accuracy remains unchanged
ABOUT FIRMWARE VERSION 2.30 [Manuals version 1.1.1]
TAGARNO Firmware Version 2.30 contains the following additions and corrections:
1. New options in Settings Menu:
- Keyboard layout. International keyboard support for annotations in TMA. Supported countries (ISO 3166-1 alpha-2): AU, BE, DE, DK, ES, FI, FR, GB, HU, IL, IT, NO, PL, SE and US
- Foot switch configuration. Select between Zoom/Focus and Zoom only. Zoom/Focus is the default configuration
Setting this option to Zoom only will allow the user to zoom in/out with the foot switch, even if the microscope is set to manual focus mode by the XPLUS FHD Control box
2. For easy access to perform a WBC, this item has been moved to the top of the camera Menu. When white balance calibration is performed, the menu will exit automatically.
ABOUT FIRMWARE VERSION 2.10 [Manuals version 1.1]
TAGARNO Firmware Version 2.10 contains the following additions and corrections:
1. TAGARNO Measurement Application*
2. Keyboard support. This firmware version contains keyboard support, so that you can connect and use a keyboard in applications
3. New options in the setup menu
- A manual white balance calibration (WBC) option has been added
- Enabling/disabling white balance calibration (WBC), when the LED light is switched on or off
- Change the frame rate between video formats 1080p60/1080p50/1080p30/1080p25
(1080p30/1080p25 is not recommended but can be used to perform USB 3.0 video capture on a PC which does not meet the minimum requirements)
- The model number has been removed from the System information window
*Access to TAGARNO Measurement Application requires license key
to order TAGARNO Measurement Application and retrieve the activation code to request a unique license key
DOWNLOAD LATEST TAGARNO FIRMWARE VERSION
DOWNLOAD LATEST TAGARNO FIRMWARE VERSION